Please preference Specification and Pricelist
FEATURES / TÍNH NĂNG
Features:
 3 LCR Measurement Functions
 1% Basic Accuracy
 Test Frequency 120Hz/1kHz and Equivalent Circuit depends on selected Measuring Function & Range
 High Resolution: 0.1uH, 0.1pF and 1mΩ
 Dissipation Factor Measurement
 Zero Adjustment
 Optional Accessory: TL06 for SMD Type Component
Enclosed within a sturdy shell, LCR814 Hand Held LCR meter is a reliable and easytouse workhorse for quick capacitance, inductance, and resistance measurement. Large 0.5", 3 1/2 digits display guarantees clear view in typical yet less than perfect conditions such as field services. But convenience is not the only virtue; with 0.1 H/0.1pF/1m high resolutions, dissipation factor measurement, and zero adjustment for removing lead impedance, LCR814 has the capability in precision to almost replicate that in true incircuit measurements.
SPECIFICATION / THÔNG SỐ KỸ THUẬTCAPACITANCE  
Range  200pF, 2nF, 20nF, 200nF, 2uF, 20uF, 200uF, 2mF, 20mF in 9 ranges  
Accuracy  200pF~200uF +(1% rdg + 2 digits), 2mF~20mF +(2% rdg + 10 digits) 

Dissipation Factor  Range: 0~1.999  
Accuracy: 2nF < Cx < 2uF + (1% + 10 + 2000 / Cx) 2uF < Cx < 2mF + (1% + 20 + 2000 / Cx) 200pF, 20mF not specified 

INDUCTANCE  
Range  200uH, 2mH, 20mH, 200mH, 2H, 20H, 200H in 7 ranges  
Accuracy  200uH, 2H~20H+(2% rdg + 2 digits) 2mH~200mH+(1% rdg + 2 digits) 200H+(3%rdg + 2 digits) 

Dissipation  Range 0~1.999  
Factor  Accuracy: Lx + 200mH + (1% + 10 + 2000 / Lx) 200mH~200H + (2% + 20 + 2000 / Lx) 

RESISTANCE  
Range  2Ω, 20Ω ,200Ω , 2kΩ , 20kΩ , 200kΩ ,2MΩ, 20MΩ in 8 ranges  
Accuracy  2Ω +(1% rdg+ 5 digits) 20Ω~200kΩ +(1% rdg+ 2 digits) 2MΩ~20MΩ +(2% rdg+ 2 digits) 

OTHERS  
Low Battery Warning  Shows "LO BAT" when the battery level reaches 5%  
Current Consumption  10mA  
Temperature  Operation 0°C ~ 40°C, Storage 20°C ~ 70°C  
DISPLAY  
0.5' digital height, 3 1/2 digits LCD  
POWER SOURCE  
Single 9V 006P battery  
DIMENSIONS & WEIGHT  
88(W) × 40(H) × 177(D) mm Approx. 400g 